We at Piercers.com wanted you to know that SciCan recommends a separate surge protector for your Statim and Bravo Autoclaves and Hydrim Washers.
SciCan Technical Service Bulletin
Bulletin number 18-TSB-387
- Europe & International
Description: Canada, Europe & International markets (not USA); Re: All HYDRIM® Washer Disinfector units, C61WD and M2 G4—Surge Protector Recommendation.
This technical service bulletin is to recommend that a surge protector is used for the power connection when installing HYDRIM C61WD and M2 G4 models. This is particularly important and highly recommended in areas where the voltage fluctuates in order to prevent damage to the unit’s internal components and maintain its functionality.
HYDRIM models (rating listed below) are equipped with the power cord that is specific to the region it is sold into. Please refer to the appropriate unit’s operators manual for further details on the electrical requirements. You can use a secondary type surge protectors or power strips for this purpose. Please make sure not to connect any other equipment on the same outlet/power bar.
HYDRIM M2 G4 – 230-240 VAC, 50Hz, 13A & 230-240 VAC, 60Hz, 13A
HYDRIM C61 – 200-230 VAC, 50Hz, 12A & 208-240 VAC, 60Hz, 12A
If you have any additional questions please contact your local service department or SciCan representative.
SciCan Product Management